Please use this identifier to cite or link to this item:
|Title:||Dust Impact Monitor (SESAME-DIM) on board Rosetta/Philae: Millimetric particle flux at comet 67P/Churyumov-Gerasimenko||Authors:||Hirn, Attila
DELLA CORTE, VINCENZO
Seidensticker, Klaus J.
|Issue Date:||2016||Journal:||ASTRONOMY & ASTROPHYSICS||Number:||591||First Page:||A93||Abstract:||Context. The Philae lander of the Rosetta mission, aimed at the in situ investigation of comet 67P/Churyumov-Gerasimenko, was deployed to the surface of the comet nucleus on 12 November 2014 at 2.99 AU heliocentric distance. The Dust Impact Monitor (DIM) as part of the Surface Electric Sounding and Acoustic Monitoring Experiment (SESAME) on the lander employed piezoelectric detectors to detect the submillimetre- and millimetre-sized dust and ice particles emitted from the nucleus. <BR /> Aims: We determine the upper limit of the ambient flux of particles in the measurement range of DIM based on the measurements performed with the instrument during Philae's descent to its nominal landing site Agilkia at distances of about 22 km, 18 km, and 5 km from the nucleus barycentre and at the final landing site Abydos. <BR /> Methods: The geometric factor of the DIM sensor was calculated assuming an isotropic ambient flux of the submillimetre- and millimetre-sized particles. For the measurement intervals when no particles were detected the maximum true impact rate was calculated by assuming Poisson distribution of the impacts, and it was given as the detection limit at a 95% confidence level. The shading by the comet environment at Abydos was estimated by simulating the pattern of illumination on Philae and consequently the topography around the lander. <BR /> Results: Based on measurements performed with DIM, the upper limit of the flux of particles in the measurement range of the instrument was of the order of 10<SUP>-8</SUP>-10<SUP>-7</SUP> m<SUP>-2</SUP> s<SUP>-1</SUP> sr<SUP>-1</SUP> during descent. The upper limit of the ambient flux of the submillimetre- and millimetre-sized dust and ice particles at Abydos was estimated to be 1.6 × 10<SUP>-9</SUP> m<SUP>-2</SUP> s<SUP>-1</SUP> sr<SUP>-1</SUP> on 13 and 14 November 2014. A correction factor of roughly 1/3 for the field of view of the sensors was calculated based on an analysis of the pattern of illumination on Philae. <BR /> Conclusions: Considering particle speeds below escape velocity, the upper limit for the volume density of particles in the measurement range of DIM was constrained to 10<SUP>-11</SUP> m<SUP>-3</SUP>-10<SUP>-12</SUP> m<SUP>-3</SUP>. Results of the calculations performed with the GIPSI tool on the expected particle fluxes during the descent of Philae were compatible with the non-detection of compact particles by the DIM instrument.||URI:||http://hdl.handle.net/20.500.12386/26466||URL:||https://www.aanda.org/articles/aa/abs/2016/07/aa28370-16/aa28370-16.html||ISSN:||0004-6361||DOI:||10.1051/0004-6361/201628370||Bibcode ADS:||2016A&A...591A..93H||Fulltext:||open|
|Appears in Collections:||1.01 Articoli in rivista|
Show full item record
Files in This Item:
|aa28370-16.pdf||Pdf editoriale||1.38 MB||Adobe PDF||View/Open|
checked on Jun 16, 2021
checked on Jun 16, 2021
Items in DSpace are published in Open Access, unless otherwise indicated.